Product price, specification, warranty and availability subject to change without notice. Channels , , and are referenced-single ended. Specify [0, 4, 5] to indicate channels 0, 4, and 5. Note that there are no spaces in the line. If the vectors specified in these fields are shorter than the channel vector then the value of the last element in the vector will be used for the rest of the channels.
|Date Added:||18 July 2013|
|File Size:||60.9 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
For example for cards that have 32 analog input channels, and if you want to use the maximum number of differential inputs i. Finally, set the Initial analog outputs and Final analog outputs to the desired voltage values.
Stock Quick Quote. The options and their values are not case sensitive for the National Instruments boards. Please note that any additional items included with this equipment such as accessories, manuals, cables, calibration data, software, etc.
Ni DAQCard 6062E National Instruments 12-bit PCMCIA DAQ Card
Note that the PWM signal period is the frequency inverted. Differential for the first four channels, referenced single-ended for the next four channels. We know there are plenty of differences when it comes to used equipment and quite often, choosing between different pieces is difficult, especially when the equipment is not sitting right in front of you.
Set the PWM output mode field to the desired output mode. Mercury Drive Champaign, IL All of the options follow the form:. All of the analog inputs must have the same input range for this card, so only a single scalar in each field is necessary to set the range for all the specified channels.
To set the digital 60062e values when the model is loaded or unloaded, set the Initial digital outputs and Final digital outputs to the desired values respectively.
Ni Daqcarde PCMCIA Multifunction DAQ National Instruments *tested* | eBay
Therefore, the acceptable value for duty cycle is related to the frequency at which the signal is being generated. InstraView Request Form To get started: For example, enter 4: Then set the PWM output frequency in Hz or duty cycle field to the desired frequency or duty cycle. It’s like having the store dqq to you! Attempting to do so would produce an error message. Then the ninth differential input’s positive line must be connected to channel 16, and negative line to channel For example, enter 2: Please contact us to complete 606e request.
Set the Encoder input channels field to all the encoder channels that will be used on the board for the current diagram. Hence, a scalar value will apply to all channels specified in the Analog output channels field.
NI DAQCarde :: QUARC Data Acquisition Card Support
For example, enter [0 1 66062e in the Analog input channels field and [0 0 2] in the Analog input configurations field, to set channel 0 to referenced single-ended, channel 1 to referenced single-ended, and channel 2 to differential.
For instance, channels, and are differential. Specify [0, 4, 5] to indicate that lines 0, 4 and 5 are to be configured as digital outputs.
For devices with 8 channels: We’ll send you an email letting you know exactly when your piece of equipment will be available for viewing Name: In order to have analog outputs set to a particular voltage when the model is loaded or unloaded, the analog outputs must be configured on the HIL Initialize block’s Analog Outputs tab. As mentioned earlier in this section, the counters cannot be used as both encoder inputs and PWM outputs. Specify [0, 4, 5] to indicate that lines 0, 4 and 5 are to be configured as digital inputs.
The following are the available options and valid values for this board. Next set the analog input maximum and minimum.
The supported analog input configurations are Referenced singled-ended 0Non-referenced singled-ended 1Differential 2.